Information in this article applies to:
Article ID: NDS1027 — Created: 3 Apr 2017 — Reviewed: 5 Aug 2017
You may observe that the image captured by the objective camera is not ideal and may have one or more of the following issues:
There may be multiple causes for these types of problem:
Fortunately, these issues are all easy to solve by recalibrating the objective camera and verifying the applied filter.
To correct the objective image issues:
At this point, you should load a slide into the uScope and review the objective image.
It is important to note that you may need to re-calibrate the objective camera from time-to-time. Once the camera is calibrated and color corrected, you should rarely need to re-calibrate.
Refer to the following knowledge base article to learn more about flat-field correction.
Thanks for subscribing to this knowledge base article.
You will receive an email confirmation with a confirmation link. You must click the link to receive update notifications for this knowledge base article.
You are already subscribed to this knowledge base article.
It looks like you are already subscribed to this knowledge base article.